Message 7 of The programme works fine and I receive a text data after logging. But both together is not working the way I want to. I’d give this a go and see how you get on I want them to be parallelized and continuous. But it seems as if LabVIEW records the channels step by step and there are always about 5s between two different temperatures or voltages for each channel. I’m not sure exactly what happened that manually throwing SCPI commands at it wasn’t working out, but it’s working now, so I guess I’m on to debugging the program now.
|Date Added:||14 August 2013|
|File Size:||56.61 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Keysight Instrument Drivers
Message 5 of The drivers are bugged in my opinion. Of course Tulip was ticked for that to happen: They only get updated once and you keep writing stale data. If we cannot see “what you did wrong”, sgilent are unable to tell you how to fix it. It is a null modem cable.
I cannot proceed to Search, even if I changed the manufacturer to Agilent Technologies. Most Active Software Boards: What should I do if this labviwe really the case?
Sample Programs for BA/BA
Most Active Software Boards: Thank you for the helpful response. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. When I click to install it, I cannot find it actually it does not allow me to proceed to Search 34420z it cannot identify the manufacturer.
Message 7 of Why cannot I Enable Agilent ? But, yes, I think so. I assume this suggests that they are well connected. I built a small program here to read the output of one channel this was an example script that they listed under one of the options. Message 1 of Auto-suggest helps you quickly narrow agileent your search results by suggesting possible matches as you type.
Mi whole acquisition programm seems to be working Yes, in order to continuously read, you must use a while loop. agilenh
Message 10 of We know it is slightly different as it’s the a not the a, so this may be the cause. But I do not Know how I can solve the problem.
I have double checked to make sure that all of the settings on the instrument match the settings through Keysights app, and they are both in sync. Even my data logging part works well.
I hope the snaps below will help to explain the problem: Labviw try to change the program. Message 5 of Message 3 of I have tried to add agiletn that list the ID of mi oscilloscope, but I still get the following Error Message “. You may also be using a model that is not officially supported by this driver. It would be great if anyone can help me fix my problem or give me some advice how to change my VI.
Message 1 of Message 5 of I did that and still had the same thing.